Evaluation of Microleakage and Degree of Conversion of Three Composite Res-ins Polymerized at Different Power Densities
Objectives: This study aims to evaluate the degree of conversion (DC) and microlekage scores of three different composite resins polymerized with a LED curing device in standard and extra-power mode. Material and Methods: One bulk-fill (Tetric EvoCeram Bulk-Fill – TECBF) and two conventional composite materials (Clearfil Majesty Posterior – CMP and Tetric EvoCeram – TEC) were evaluated. A total of 30 specimens were prepared for six groups (N = 5). These groups were polymerized with a LED curing device as follows: TECBF-6: 3200mW/cm2 for six seconds, TECBF-20: 1000mW/cm2 for 20 seconds, CMP-6: 3200mW/cm2 for six seconds, CMP-20: 1000mW/cm2 for 20 seconds, TEC-6: 3200mW/cm2 for six seconds, TEC - 20: 1000 mW/cm2 for 20 seconds. After 24 hours of water storage, DC was measured by Raman spectroscopy. Microleakage scores of the six groups were bonded to various adhesive systems (Clearfil SE Bond or Adhese Bond Universal) were also evaluated at Class II box cavities (N = 10). Results: While the highest DC was found at the top (TECBF-20= 79.92% and TECBF-6= 79.02%) and bottom surfaces (TECBF-20 = 68.94% and TECBF-6= 71.04%) for TECBF groups, TEC groups (TEC-20top = 59.06%, TEC-6top=49.66%, TEC-20bottom = 43.72% and TEC-6bottom= 40.68%) showed the lowest DC for the both surfaces (p < 0.05). Polymerization of materials in standard or extra-power mode was similar to DC (p > 0.05). Microleakage scores were found to be similar (p > 0.05). Conclusion: According to the results of the study, different power densities of LED curing light did not affect the DC of composite resins and microleakage values of restorations at small Class II cavities.
Bulk-fill composite; degree of conversion; LED curing device; microleakage; Raman spectroscopy.
All manuscript's copyright ownership of the article "____________________________________________________________________" is transferred from the author(s) to the BRAZILIAN DENTAL SCIENCE, in the event the work is published. The manuscript has not been published elsewhere and that it has not been submitted simultaneously for publication elsewhere.
We hereby attest the study is original and does not present manipulated data, fraud or plagiarism. We made significant scientific contribution to the study and we are aware of the presented data and in agreement with the final version of the manuscript. We assume complete responsibility for the ethical aspects of the study.
This text must be printed and signed by all authors. The scanned version should be submitted as supplemental file during the submission process.