Evaluation of fracture resistance of zirconia monolithic restorations: an in vitro study
Objective: The aim of the study was to evaluate fracture resistance of zirconia monolithic restorations after ageing procedures. Material and methods: The fracture resistance of zirconia-based restoration after ageing procedures was evaluated. Monolithic translucent zirconia 3-unit FDPs were fabricated using Cerec inLab CAD/CAM system on 2 stainless steel dies prepared from stainless steel with a uniform 120 degrees circumferential deep chamfer finish line of 1 mm width and were placed into a stainless steel holder. FDPs were divided into 2 groups, the first group (Group A) was subjected to aging procedures in an autoclave at hydrothermal conditions 134 °C /2 bars for 5 hours. The second group (Group B) was not subjected to any ageing procedures and was designed as control group. All specimens of group A (Aged TZI) and group B (non- aged TZI) were loaded compressively in a universal testing machine at cross head speed 0.5 mm/min until fracture occurred. The percentage of monoclinic (m) phase was detected by XRD device. Scanning electron microscope (SEM) was used to examine the fractured surfaces for Aged TZI (Group A) and Non-aged TZI (Group B). Student’s t-test was used to compare between fracture resistances of both groups. The significance level was set at P ? 0.05. Results: The fracture resistance mean (SD) values of non-aged TZI (Group B) was 2406.9±306.8 N which showed statistically significantly higher than that of aged group (Group A) which was 1964.5±234.5 N. The percentage of monoclinic (m) phase detected by XRD device software in non-aged TZI (Group B) was nearly 0 weight % and in aged TZI (Group A) was about 42 weight %. Conclusion: Accelerated artificial aging decreases the fracture resistance of monolithic TZI FDPs.
Artificial aging; Fracture resistance; Monolithic zirconia.
Raigrodski AJ. Contemporary materials and technologies for all ceramic fixed partial dentures: a review of literature. J Prosthet Dent 2004;92:557-62.
Al-Amleh B, Lyons K, Swain M. Clinical trial in zirconia: A systemic review. J Oral Rehabil 2010;37:641-52.
Luthy H, Filser F, Loeffel O, Schumacher M, Gauckler L, Hammerle C. Strength and reliability of four-unit all-ceramic posterior bridges. Dent Mater 2005;21:930-7.
Beuer F, Schweiger J, Eichberger M, Kappert HF, Gernet W, Edelhoff D. High strength CAD/CAM-fabricated veneering material sintered to zirconia copings- a new fabrication made for all-ceramic restorations. Dent Mater 2009;25:121-8.
Liu Y, Feny H, Bao Y, Qiu Y, Xing N., Shen Z. Fracture and interfacial delamination origins og bilayer ceramic composites for dental restorations. J Eur Ceram Soc 2010;30:1297-307.
Denry I, Kelly JR. State of the art of zirconia for dental applications. Dent Mater 2008;24:299-307.
Zarone F, Russo S, Sorrentino R. From porcelain-fused-to-metal to zirconia: clinical and expermintal considerations. Dent Mater 2011;27(1):83-96.
Miyazaki T, Nakamura T, Matsumura H, Ban S, Kobayashi T. Current status of zirconia restoration. J Prosthodont 2013;57:236-261.
Liu PR, Essig ME. Panorama of dental CAD/CAM restorative systems. Compend Contin Educ Dent 2008;29:482-8.
Egilmez F, Ergun , Cekic-Nagas I, Vallittue PK, Lassila LV. Factors affecting the mechanical behavior of Y-TZP. J Mech Behav Biomed Mater 2014;37:78-87.
Garvie RC, Hannink RH, Pascoe RT. Ceramic steel? Nature 1975;258:703-704.
Lughi V, Sergo V. Low temperature degradation-aging-of zirconia: A critical review of the revelent aspects in dentistry. Dent Mater 2010;26(8)807-20.
Tinschert J, Zwez D, Marx R, Anusavice KJ. Structural reliability of Alumina-,Feldspar-, leucite-, mica-and zirconia-based ceramics. J Dent 2000;28:529-35.
Teixeira EC, Piascik JR, Stoner BR, Thompson JY. Dynamic fatigue and strength characterization of three ceramic materials. J Mater Sci Mater Med 2007;18:1219-24.
Pittyayachawan P, McDonald A, Petrie A, Knowles JC. The biaxial flexural strength and fatigue property of Lava Y-TZP dental ceramics. Dent Mater 2007;23:1018-29.
Kelly JR, Denry I. Stabilized zirconia as a structural ceramic: an overview. Dent Mater 2008;24:289-98.
Guo X. Property degradation of tetragonal zirconia induced by low temperature defect reaction with water molecules. Chem Mater 2004;16:3988-3994.
Yashimura M, Noma T, Kawabata K, Somiya S. Role of H2O on the degradation process of Y-TZP. J Mater Sci Lett 1987;6:465-467.
Kim HT, Han JS, Yang JH, Lee JB, Kim SH. The effect of low temperature aging on the mechanical property and phase stability of Y-TZP ceramics. J Adv Prothodont 2009;1:113-117.
Flinn BD, Raigrodski AJ, Singh A, Mancl LA. Effect of hydrothermal degradation on three types of zirconias for dental application. J Prosthet Dent 2014;112:1377-1384.
Flinn BD, deGroot DA, Mancl LA, Raigrodski AJ. Accelerated aging characteristics of three ytrria-stabilized tetragonal zirconia polycrystalline dental materials. J Prosthet Dent 2012;108:223-30.
Hip implants recalled because of potential fracture problems, FDA patient safety news: show #2, March 2002. Available at: http://www.accessdata.fda.gov/scripts/cdrh/cfdocs/psn/printer.Cfm?id=107.Accessed September 4, 2014.
Kobayashi K, Kuwajima H, Maski T. Phase change and mechanical properties of ZrO2-Y2O3 solid electrolyte after aging. Solid state Ionics. 1981;3:489-93.
Lange FF, Dunlop GL, Davis BL. Degradation during aging of transformation-toughened ZrO2-Y2O3 materials at 250 ºC. J Am Ceram Soc 1986;69:237-40.
Kvam K., Karlsson S. Solubility and strength of zirconia-based dental materials after artificial aging. J Prothet Dent 2013;110:281-87.
Gaillard Y, Jiménez-Piqué E, Soldera F, Mücklich F, Anglada M. Quantification of hydrothermal degradation in zirconia by nanoindentation. Acta Mater 2008;56:4206-16.
Kohorst P, Dittmer MP, Borchers L, Stiesch-Scholz M. Influence of cyclic fatigue in water on the load-bearing capacity of dental bridges made of zirconia. Acta Biomater 2008;4:1440-4.
Kitano Y, Iwanaga H, Oishi M, Mutsuda Y, Masaki T. Structural and mechanical changes of zirconia and zirconia composites after aging. In: Duran P, Fernandz JF, editors. 3rd euro-ceramics, Vol.3: engineering ceramics. Spain: Faenza Editrice Iberica S.L; 1993.p. 647-52.
Chevalier J, Cales B, Drouin JM. Low temperature aging of Y-TZP ceramics. J Am Ceram Soc 1999;82:2150-4.
Ruhle M, Claussen N, Heuer AH. Microstructural studies of Y2O3-containing tetragonal ZrO2 polycrytals (Y-TZP). In: Claussen N., Ruhle M., Heuer AH., editors. Advances in ceramics, Vol. 12, Science and technology of zirconia ZI. Columbus, OH: The American Ceramic Society, Inc.: 1984.P.352-70.
Sato T, Ohtaki S, Endo T, Shimada M. "a changes in crystallographic phase and microstructure on the surface of the yttria-doped tetragonal zirconia poly-crystals (Y-TZP) by annealing in humid condition." – Advances in ceramics, 24, Science and Technology of zirconia III, ed. Somiya S., Yamamoto N., Yanagida H.: The American Ceramic Society Columbus, 1988;501-8.
Sato T, Shimada M. Transformation of ytrria-doped tetragonal ZrO2 polycrystal by annealing in water. J Am Ceram Soc. 1985;68:356-359.
Cattani-Lorente M, Scherrer SS, Ammann P, Jobin M, Wiskott HW. Low temperature degradation of a Y-TZP dental ceramic. Acta Biomater 2011;7:858-865.
Kosmac T, Oblak C, Jevnikar P, Funduk N, Marion L. The effect of surface grinding and sandblasting on flexural strength and reliability of Y-TZP zirconia ceramic. Dent Mater 1999;15:426-33.
Chevalier J, Gremillard L, Deville S. Low temperature degradation of zirconia and implications for biomedical implants. Annu Rev Mater Res 2007;37:1-32.
Deville S, Gremillard L, Chevalier J, Fantozzi G. A critical comparison of methods for the determination of the aging sensitivity in biomedical grade yttria-stabilized zirconia. J Biomed Mater Res B Appl Biomater 2005;72(2):239-45.
Pereira GK, Venturini AB, Silvestri T, Dapieve KS, Montagner AF, Soares FZ, Valandro LF. Low temperature degradation of Y-TZP ceramics: A systematic review and meta-analysis. J Mech Behav Biomed Mater 2015;55:151-63.
Amaral M, Valandro LF, Bottino MA, Souza RO. Low temperature degradation of Y-TZP ceramic after surface treatments. J Biomed Mater Res B: Appl Biomater 2013;101:1387-92.
International Organization for Standardization ISO 13356: 2008 Implants for surgery—Ceramic materials based on yttria-stabilized tetragonal zirconia (Y-TZP). Geneva: International Organization for Standardization. 2008 Available at: http://www.iso.ch/iso/en/prods-services/ISOstore/store.html.
Kim JW, Covel NS, Guess PC, Rekow ED, Zhang Y. Concerns of hydrothermal degradation in CAD/CAM zirconia. J Dent Res 2010;89:91-5.
Tanaka K, Tamura J, Kawanabe K, Nawa M, Uchida M, Kokobo T, Nakamura T. Phase stability after aging and its influence on pin-on-disk wear properties of Ce-TZP/Al2O3 nanocomposite and conventional Y-TZP. J Biomed Mater Res A. 2003;67:200-7.
Tanaka K, Tamura J, Kawanabe K, Nawa M, Oka M, Uchida M, Kokubo T, Nakamura T. Ce-TZP/Al2O3 nanocomopsite as a bearing material in total joint replacement. J Biomed Mater Res B 2002;63:262-70.
Ban S, Sata H, Suehiro Y, Nakanishi H, Nawa M. Biaxial flexure strength and low temperature degradation of Ce-TZP/Al2O3 nanocomposite and Y-TZP as dental restoratives. J Biomed Mater Res B Appl Biomater 2008;87(2):492-8.
Algazzawi T, Lemons J, Liu PR, Essig M, Bartolucci A, Janowski G. Influence of low-temperature environment exposure on the mechanical properties and structural stability of dental zirconia. J Prosthodont 2012;21:363-9.
Siarampi E, Kontonasaki E, Andrikopoulos KS, Kantiranis N, Voyiatzis GA, Zobra T, Paraskevopoulos KM, Koidis P. Effect of in vitro aging on the flexural strength and probability to fracture of Y-TZP zirconia ceramics for all- ceramic restorations. Dent Mater 2011;30:306-16.
Nakamura K, Harada A, Kanno T, Inagaki R, Niwano Y, Milleding P, Örtengren U. The influence of low-temperature degradation and cyclic loading on the fracture resistance of monolithic zirconia molar crowns. J Mech Behav Biomed Mater 2015;47:49-56.
Cattani-Lorente M, Durual S, Amez-Droz M, Wiskott HW, Scherrer SS. Hydrothermal degeneration of a 3 Y-TZP translucent dental ceramic: A comparison of numerical predication with experimental data after 2 years of aging. Dent Mater 2016;32:394-402.
Raigrodski AJ. Contemporary all-ceramic fixed partial dentures: a review. Dent Clin North Am 2004;48:531-544.
Sundh A, Molin M, Sjogren G. Fracture resistance of yttrium oxide partially-stabilized zirconia all-ceramic bridges after veneering and mechanical fatigue testing. Dent Mater 2005;21:476-82.
Choi YS, Kim SH, Lee JB, Han JS, Yeo IS. In vitro evaluation of fracture strength of zirconia restoration veneered with various ceramic materials. J Adv Prosthodont 2012;4(3):162-9.
Rosentritt M, Plein T, Kolbeck C, Behr M, Handel G. In vitro fracture force and marginal adaptation of ceramic crowns fixed on natural and artificial teeth. Int J Prosthdont 2000;13:387-91.
Manicone PF, Rossi Iommetti P, Raffadllo L. An overview of zirconia ceramics: Basic properties and clinical applications. J Dent 2007;35:819-26.
Shah K, Holloway JP, Denry IL. Effect of coloring with various metal oxide on microstructure, color, and flexural strength of 3Y-TZP. J Biomed Mater Res B Appl Biomater 2008;87:329-37.
Li D, Liu W, Zhao S, Feng H, Yang D, Gao S. Effect of cerium oxide and erbium oxide as colorants on the chromaticity and mechanical properties of dental zirconia ceramic. J Clin Rehabil Tissue Eng Res 2011;47:8829-32.
Tortopidis D, Lyons MF, Baxendale RH, Gilmour WH. The variability of bite force measurement between sessions in different positions within the dental arch. J Oral Rehabil 1998;25-681-6.
Swain MV, Rose LRF. Limitations of transformation-toughened zirconia alloys. J AM Ceram Soc 1986;69:911-8.
Chevalier J. What future for zirconia as a biomaterial? Biomaterials 2006;27:535-43.